What does a ‘Comm Error See Manual’ message mean?
This message is displayed on Inogen oxygen concentrators.The concentrator is producing oxygen but cannot report the battery status. Replace the battery. If the message persists the concentrator may need a service.
This is a low-priority alert message and is accompanied by a double beep and a solid red light.